|Place of Origin:||China|
|Certification:||TUV CE, ISO13485, ISO9001|
|Model Number:||ITI Straumann TissueLevel Synocta®|
|Minimum Order Quantity:||10Pcs|
|Price:||USD8~12/pc, included screw|
|Delivery Time:||3~5 days|
|Payment Terms:||T/T, Western Union, MoneyGram|
|Material:||Castable Plastic||Warranty:||12 Months|
|After-sale Service:||Online Technical Support||Connection:||Engaging & Non-Engaging|
|Size:||RN 4.8mm/ WN 6.5mm||Properties:||Implant Materials & Artificial Organs|
TUV CE UCLA Abutment,
Castable Plastic UCLA Abutment,
6.5mm UCLA Implant Abutment
TUV CE WN 6.5mm Castable Plastic UCLA Implant Abutment
ITI Straumann Tissue Level Synocta® UCLA CoCr Base Castable Abutment Compatibe
RN 4.8mm/ WN 6.5mm (Engaging & Non-Engaging)
CoCr base and castable plastic chimney
Made of CoCr base and castable plastic chimney material for the stable retention of modelling materials
This base is available in two types of platforms: engaging and non-engaging.
Approved by CE, ISO 9001, ISO 13485
Universal Clearance-Limited Abutments (UCLAs) are indicated for laboratory use to manually create an implant-level custom abutment for a cement- or screw-retained restoration. UCLAs are precisely machined and attahed to the implant fixture (or implant analog) with a titanium screw of guide pin. The plastic sleeve on top of the abutment provides a supporting structure on which to wax the restoration. Each UCLA is specific to the restorative platform of the corresponding implant.
Plastic UCLAs are used to create diagnostic wax-ups (try-in prostheses).
These engaging UCLAs are indicated for single-unit resotrations to prevent rotation.
Each Inclusive UCLA is packaged with a guide pin (Inclusive Guide Pin) and separate retaining screw (Inclusive Titanium Screw). The guide pin should be used during the fabrication process. The titanium screw is used while fitting the diagnostic wax-up, or for retaining a definitive restoration.
Contact between Inclusive Plastic UCLAs and soft tissue should not exceed a one (1) hour duration.
Angled abutments should not be used to restore small-diameter implants (less than or equal to 3.0 mm) in the posterior region.
Inclusive UCLAs are shipped NON-STERILE. Non-sterile items must be cleaned, disinfected, and sterilized prior to clinical use, according to a validated method.
Cleaning: Wash using a broad spectrum cleaning solution, followed by thorough rinsing and drying.
Disinfection: Immerse abutments in disinfectant1, rinse with distilled water and dry.
Sterilization: Gravity-fed sterilizers: Autoclave in sterilization pouch for 15 minutes at 132°C (270°F). Allow sterilized components to dry for at least 30 minutes.